I don't know if you've followed all my comments, but I found that the
replacement MV=95 that I bought in 2011 but didn't start using until
2015 was disappointing in its reduced sensitivity.
I urge you to try a new Arnold unit. I won't bother going back unless I
receive a free Sigma (which I doubt).
On 2/23/2017 15:55, Martin Fisher wrote:
> Dang it! My MV-95 just sprung a microscopic pinhole leak!! Aargh!!! Luckily it's in one of the lower corners and a piece of scotch type of tape is doing the job so far but I'll probably be on the prowl for another one soon.
Richard L. Hess email: [log in to unmask]
Aurora, Ontario, Canada 647 479 2800
Quality tape transfers -- even from hard-to-play tapes.